A short paper in Optics Express looks interesting. In A high speed wavefront determination method based on spatial frequency modulations for focusing light through random scattering media, Meng Cui presents a method for rapidly determining the optimal wavefront to ‘cancel out’ the scattering when 785nm light passes through turbid media. In his example, a glass diffuser was used, but the clear goal for this work is to replace the glass with a brain.
To understand why this is so important for in vivo two-photon imaging, let’s review how 2-p imaging works. Light from a laser is focused to a point and swept across the field in a raster. The resulting fluorescence is of a different wavelength and can thus be filtered out from the excitation light. For each voxel, all the fluorescence that re-enters the objective is collected, regardless of its source. The total amount of fluorescence collected for that timepoint in the sweep is assigned as the brightness of that voxel. Since the user knows where the laser was being aimed, scattering of fluorescence emission may reduce the brightness but will not blur the image. However, scattering of the excitation light can dramatically reduce the excitation at the target voxel while increasing the off-target excitation of its neighbors. This causes a rapid increase in background fluorescence and blur at increasing brain depth.
Previous reports work has shown that one can use adaptive optics to adjust the phase of the wavefront of the excitation light to correct for the scattering of the excitation. However, determination of the optimal wavefront for a field of view took minutes, which could be problematic for imaging in an awake animal. Any changes in the precise position of the brain might change the optimal wavefront. Ideally, one would want a system that could optimize the wavefront every second, or even before every frame of acquisition (typically 4-8 Hz in a raster scan in vivo experiment)
I’ll let Meng Cui explain the technique in his own words
Elastic scattering is the dominant factor limiting the optical imaging depth in tissues. Take gray matter as an example, at 800 nm the scattering coefficient is 77 /cm and the absorption coefficient is 0.2 / cm. If there is a way to suppress scattering, the optical imaging depth could be greatly improved. Despite the apparent randomness, scattering is a deterministic process. A properly engineered wave can propagate inside scattering media and form a focus, a well understood phenomenon in the time reversal and optical phase conjugation (OPC) studies…
For applications on biological tissues, acquisition time on the order of one millisecond (ms) per degree of freedom is desired. Deformable mirrors can provide a high modulation speed. However the degrees of freedom are rather limited. A phase-only SLM can provide about one million degrees of freedom at a much lower modulation speed. In this work, I present a novel method, capable of providing as many degrees of freedom as a SLM with a data acquisition time of one ms per degree of freedom. The method was employed to focus light through a random scattering medium with a 400 ms total data acquisition time, ~three orders of magnitude faster than the previous report .
The essence of a COAT system is to phase modulate different input spatial modes while detecting the output signal from the target. To greatly improve the operation speed, the experiment requires a device that can provide fast phase modulation and can access a large number of spatial modes very quickly. To meet these two requirements, a pair of scanning Galvanometer mirrors was used to quickly visit different modes in the spatial frequency domain or k space, and a frequency shifted reference beam was provided for a heterodyne detection. The wavefront profile was first determined in k space and then transformed to the spatial domain. The spatial phase profile was displayed on a SLM to focus light onto the target. In such a design, the number of degrees of freedom is limited by the number of pixels on the SLM and the experiment speed is determined by the scanning mirror speed…
Compared to existing techniques, the reported method can provide both a high operation speed and a large number of degrees of freedom. In the current design, the operation speed is limited by the scanning mirror speed and the maximum number of degrees of freedom is limited by the SLM pixel number. In this demonstration, 400 spatial modes in k space were visited and the determined phase profile was displayed on the SLM. Depending on the scattering property of the media, more (up to 1920 x 1080) or less number of degrees of freedom can be used to optimize the focus quality and the operation speed.
Using a stepwise position scanning, the method achieves an operation speed of one ms (400 μs transition time + 600 μs recording time) per spatial mode, ~three orders of magnitude faster than the previous report. Using a continuous position scanning and a faster position scanner such as resonant scanning mirrors, polygon mirror scanners, or acousto-optic deflectors, the operation speed can be potentially increased by at least one order of magnitude. It is anticipated that the reported technique will find a broad range of applications in biomedical deep tissue imaging.